This product uses “real-time fluorescent quantitative PCR technology” combined with self-developed “real-time dynamic precision temperature control technology”, which can be used in various fluorescent quantitative PCR detection projects to achieve accurate detection. rapid nucleic acid With powerful features and flexible operation, DNA and RNA PCR amplification and detection experiments can be completed in a compact and portable environment.
Using a special reverse anti-pollution open cap design, can effectively prevent contamination caused by accidental bursting of the hot cap, simplify the experimental operation process, and combine with efficient and fast PCR reagents to provide a complete solution for SARS. -CoV- 2 detection and other projects.
1. Equipment Features
- ORF1ab and N gene
- RNase P as an internal control
- CT cutoff: ≤30
- LoD: 500 copies/mL
- Throat smear or sputum
- This kit can also be compatible with ABI7500, BIORAD CFX96, AGS4800
2. Operation Steps (Combine with Nucleic Acid Releasing Reagent DA0940)
- Sampling and inactivation
- Rock and rest for 5 min, then centrifuge
- Add the extracted sample to the PCR tubes with PCR reagent
- Perform PCR, within 0.5 hours to obtain the result
The following basic safety measures should be observed during the operation, maintenance and repair of the instrument. If the notes, warnings or cautions specified in the manual are neglected, the basic protection provided by the instrument may be affected. Meanwhile, it may also damage the security level of the design and manufacture of instruments and the intended scope of application.
It is not allowed to change the parts of the instrument without permission, and the inspection and maintenance of the instrument must be completed by engineers licensed by Hangzhou AnYu Technologies Co., Ltd. If the equipment is not used in the manner specified in this manual, the protection provided by the equipment may be damaged
The wells and hot cap of the instrument should be periodically cleaned with special brushes bathed in a little absolute alcohol to guarantee complete contact between tubes and wells and high thermal conductivity while free of pollution. Use a soft cloth to do daily maintenance to clean the surface and wells. If there is any stain on the surface, you can clean it with a dust-free cloth with 75% alcohol. Do not use chlorine disinfectants, moisturizers, antibacterials soaps, aromatic solutions, propanol or isopropanol to clean the equipment, as they are types of cleaning or sanitizing agent is easy to have a chemical reaction with parts of the equipment or materials contained in the equipment and cause danger.
Perform only the maintenance mentioned in this manual. Before using any cleaning and disinfection method not described in this document, please contact the technical support personnel of the Company or authorized agency to ensure that the method used will not damage the instrument. If you have any questions on the compatibility of disinfectants or cleaning agents with pieces of equipment or materials contained in the equipment, consult the Company or an authorized agency.